Florin of YourITronics did a teardown and calibration of a HP Agilent E3611A power supply: I got this HP Agilent E3611A from Ebay. It was marked as not working and I was planing a repair video on it. Unfortunately the unit turned out to be fully functional so I only did a teardown and calibration.
Florin of YourITronics writes: In this video I am calibrating the Gophert CPS-3205C power supply using my Keithley 175 4.5 digit multimeter and my adjustable dummy load. The calibration procedure is: 1. Set the V/A selector switch to A position, 2. Push the adjust knob then power on the power supply,the meter will display PUSH […]