Teardown, repair and analysis of an Agilent E4443A 3Hz – 6.7GHz PSA series Spectrum Analyzer from The Signal Path:
In this episode Shahriar repairs an Agilent PSA Series Spectrum Analyzer. The instrument generates many errors during self-alignment and produces no measurements below 3.2GHz. The block diagram of the unit is thoroughly presented and various possible failure points are considered. Based on the observation of the noise floor, the most likely cause is the second LO module. The measurement of the LO power indicates that the second LO power is fall below nominal.
More details at The Signal Path homepage.