Another video review from TheSignalPath, Keysight EXA Signal Analyzer/Spectrum analyzer: In this episode Shahriar reviews the Keysight EXA Signal Analyzer / Spectrum Analyzer (N9010A). The X-Series Signal Analyzers and Spectrum analyzers from Keysight are compared against each other. The EXA … Read more
Tag Archives: teardown
Rigol DSA875-TG 9kHz – 7.5GHz Spectrum Analyzer & tracking generator review, teardown & experiments from TheSignalPath: In this episode Shahriar takes a close look at the top-of-the-line Rigol DSA875-TG 7.5GHz Spectrum Analyzer. This unit which, offers a built in tracking generator … Read more
Steve did a teardown and refurbishment of a Cambridge Audio A4 Amplifier: In this video, Steve disassembles a Cambridge Audio A4 Integrated Amplifier which he rescued before it was thrown away. The amplifier is refurbished with new electrolytic capacitors and a … Read more
Teardown, repair and analysis of an Anritsu MS2721A 7.1GHz portable spectrum analyzer by Shahriar of TheSignalPath: In this episode Shahriar takes a look at an Anritsu MS2721A portable spectrum analyzer which does not completely boot up. After displaying the message … Read more
Review and teardown of the Analog arts SL987 USB oscilloscope AWG pattern generator and Logic analyzer
Review and teardown of the Analog arts SL987 USB oscilloscope AWG pattern generator and logic analyzer by Shahriar of TheSignalPath: In this episode Shahriar takes a close look at the Analog Arts USB based mixed-domain instrument. The SL987 offers an oscilloscope, arbitrary … Read more
Kerry Wong writes: In my last post, I showed my LED backlight mod of the HP 8642B synthesized signal generator. And during the process, I also showed some pictures of the A1 (keyboard and LCD) module and A2 (modulator) module … Read more
Teardown and repair of a Fluke 196B Handheld ScopeMeter: In this episode Shahriar goes through the teardown and repair process of a Fluke 196B Handheld ScopeMeter. This unit, which has been purchased as a “dead” unit from eBay, is examined … Read more
Kevtris tears down the Tektronix PS2521G power supply. He produced this post detailing his findings.: Weellll, I needed a good bench power supply, so I finally broke down and bought one. I got a Tektronix PS2521G. This doodad has three isolated outputs. This … Read more
Free PCB builder extraordinaire Kenneth did an extensive teardown of an HP ProCurve 2824 Ethernet Switch. He analyzes the PCB piece by piece, explaining the smaller function blocks and how they fit together. So that is how all of the internals of … Read more
Gelbanana had some free time on his hands and he decided to try and repair an old Weller DS701E soldering-desoldering station he had laying around. The problem was diagnosed to a couple of bad capacitors. After a quick replacement, the … Read more
Presslab tears down the Weller WX 2021 soldering station. He is hunting for the pinout of the front panel serial connector. He was able to identify the serial IC used, and with it the pinout. Via the forum.
Bob picked up a huge LED matrix sign on eBay that has power supply problems. After fixing the power supply, he decided to replace the controler boards with an Arduino. Via Hacked Gadgets.
Mat received his open 7400 contest prize, the Analog Discovery bench tool. He was quick to tear it down and find out how it ticks. The brain of the beast is a Xilinx Spartan-6 FPGA which works in conjunction with … Read more
Arhi tears apart the Voltcraft PS-1305/230 isolated 0-230 volts AC 4.5 Amp power supply. It also comes with a 0-30V 5A, and 5V 3A DC channels. The laboratory power unit is a combination of direct current power unit and safety … Read more
Andrew decided to write up a how-to on DIY BGA soldering. His article describes why it’s a bad idea to keep the vias unfilled with solder before reflow. Capilary force sucks up the ball of solder into the via, and … Read more
Hak8or got his hand on an X-ray bone mineral density assessment machine someone dumped. It seems the only problem is a faulty temperature sensor, but Hak80r decided to break it down and share the pictures with us. For an in-depth … Read more
Kenneth scored two HP universal frequency counters and popped them open. One generates the timebase from an uncompensated oscillator in a TO-5 package, while the other uses a large temperature controlled oven oscillator: here is both of my counters measuring … Read more
Mick bought and disassembled a Tektronix 465 oscilloscope. It’s a 100 MHz dual trace, dual timebase analog scope. He points out that most of the transistors are socketed instead of soldered. More pics below. Via the forum.
Raj has been designing a soldering station that would support Weller’s RT type tips with temperature sensor and heater built into them. A stereo-type jack on the iron connects them to the soldering station cable. So far Raj has reverse … Read more
Victor took apart a defective LCD monitor and repaired it. The LCD didn’t respond at all to the power button, so chances were good something was off in the power supply. Sure enough there was a bulging cap, replacing it … Read more